- 10MHZ bipolar radio frequency (RF) with two rollers can penetrate into the layer of 0.5-1.5 cm below the skin to work on the adipose tissue effectively.
- 700-2000nm infrared light can heat the connective tissue of accelerate the regeneration of collagen and elastic fibers. It can also improve the blood circulation and lymph circulation to promote the metabolism.
- 0-0.07MPa adjustable vacuum can suck the target area into the space between the two rollers which are actually 2 electrodes. This can make the treatment accurately and effectively. It can also make the treatment more comfortable. The auto-rollers can do massages as well
Treatment Mechanism is the technology of controllable suction electric heating.
The unit controls by the Microcomputer is composed of system console, operating Interface and treatmenthead.
The controlling system located at the main console.
The embedded Microcomputer monitors and controls the system continuously.
The embedded microprocessor monitors can control the system continually.
The operating Interface includes a full-LCD screen (8.0inch), the key switch and emergency shut-off knob.
LCD screen both shows the system work program, treatment parameters and number of light emitting.
The treatment head operated by hand comprises the intensive emitter and transmission system.
The system is designed in strict accordance with standard EN60601-1:1990+ A1:1993+ A2:1995:1996.
RF power: Up to 50W
RF frequency: 10M Hz
Infrared light power: Up to 20W
IR Spectrum: 700nm-2000nm
Negative pressure: 0-0.07MPa
Treatment Area Size: (Body): 50mmx55mm (Arm): 37mm x 23 mm
Electrical requirement: 240 VAC; 50Hz;400VA
Dimension (W x D x H): 454x390x1155mm